Selection will refresh the page with new results
Hanna GroLine Soil Test Direct Soil Conductivity Tester
Out of stock
The Hanna GroLine Soil Test Direct Soil Conductivity Tester - HI98331 is a pocket tester specifically designed to directly measure soil conductivity (EC) and temperature. With a compact size, single button operation, and automatic calibration, Soil Test is an excellent choice for taking direct conductivity measurements in soil.
Hanna GroLine Soil Test Direct Soil Conductivity TesterThe Hanna GroLine Soil Test Direct Soil Conductivity Tester - HI98331 is a pocket tester specifically designed to directly measure soil conductivity (EC) and temperature. With a compact size, single button operation, and automatic calibration, Soil Test is an excellent choice for taking direct conductivity measurements in soil.
- No need for soil slurries with the steel penetration electrode
- Integrated temperature sensor for automatic temperature compensation
- Great for environmental and agricultural testing
Lightweight & PortableMade to fit comfortably in your hand, this compact EC tester for soil has an easy-to-view LCD. The stainless steel electrode allows for direct penetration into soil and soil slurries.
Made for Any UserThe HI98331 makes testing easy for any user with single button operation and automatic calibration to 1413 µS/cm conductivity standard.
Integrated Temperature ProbeUsing a fixed 2%/C temperature correction coefficient, this EC tester has a built-in temperature sensor so that samples are automatically compensated for temperature variations.
Measure the Bulk EC of Your Soil with a Soil Conductivity TestBulk electrical conductivity of soil measures the total conductivity. Total conductivity includes the EC of the soil, air, and moisture in your sample. All these things carry charged ions that would read as EC. This reading is very useful; you can calculate your pore water conductivity and saturated extract conductivity from the result. You would need to know your water content to perform that calculation (how much water there is in your soil).
How To Measure the Bulk EC with your Direct Soil EC Meter:
- Pick your testing location.
- Rinse the testing probe with deionized water, and make sure it is dry.
- Check the soil and ensure that the soil is moist.
- Use a ruler or auger to make a hole in the soil. This keeps the testing depth consistent.
- Insert your soil conductivity tester directly into the soil, and take your measurement.
|EC Range||0 to 4000 μS/cm; 0.00 to 4.00 mS/cm (dS/m)|
|EC Resolution||1 μS/cm; 0.01 mS/cm (dS/m)|
|EC Accuracy||±50 μS/cm (0 to 2000 μS/cm); ±300 μS/cm (2000 to 4000 μS/cm); ±0.05 mS/cm (0.00 to 2.00 mS/cm); ±0.30 mS/cm (2.00 to 4.00 mS/cm)|
|EC Calibration||Automatic, one-point (1.41 mS/cm)|
|EC/TDS Temperature Compensation||Automatic, with coefficient (ȕ) fixed @ 2%/°C|
|Temperature Range||0.0 to 50.0°C (32.0 to 122.0ºF)|
|Temperature Resolution||0.1°C (0.1°F)|
|Temperature Accuracy||±1°C (±1.5°F)|
|EC/TDS Probe||114 mm (4.5 in) stainless steel penetration (fixed)|
|Automatic Shut-Off||8 minutes, 60 minutes, or can be disabled|
|Battery Type / Life||CR2032 Li-ion (included) / approx. 100 hours of continuous use|
|Environment||0 to 50°C (32 to 122°F); RH 95% max|
|Dimensions||50 x 196 x 21 mm (2.0 x 7.7 x 0.9 in)|
|Weight||74 g (2.4 oz.)|
|Ordering Information||HI98331 (Soil Test) is supplied with HI73331 penetration conductivity probe, calibration screwdriver, batteries and instructions.|
|Manufacturer Part Number||HI98331|